Table 41 to Subpart UUU of Part 63—Requirements for Installation, Operation, and Maintenance of Continuous Parameter Monitoring Systems
As stated in § 63.1572(c)(1), you shall meet each requirement in the following table that applies to you.
If you use . . .
You shall . . .
1. pH strips
Use pH strips with an accuracy of ±10 percent.
2. Colormetric tube sampling system
Use a colormetric tube sampling system with a printed numerical scale in ppmv, a standard measurement range of 1 to 10 ppmv (or 1 to 30 ppmv if applicable), and a standard deviation for measured values of no more than ±15 percent. System must include a gas detection pump and hot air probe if needed for the measurement range.
[70 FR 6966, Feb. 9, 2005]
Title 40 published on 2012-07-01
The following are only the Rules published in the Federal Register after the published date of Title 40.
For a complete list of all Rules, Proposed Rules, and Notices view the Rulemaking tab.
This is a list of United States Code sections, Statutes at Large, Public Laws, and Presidential Documents, which provide rulemaking authority for this CFR Part.