40 CFR 92.118 - Analyzer checks and calibrations.
(1) Prior to initial use and after major repairs, bench check each analyzer for compliance with the specifications of § 92.109.
(i) Leak check of the pressure side of the system (see paragraph (b) of this section). If the option described in paragraph (b)(2) of this section is used, a pressure leak check is not required.
(ii) Calibration of all analyzers (see §§ 92.119 through 92.122).
(iii) Check of the analysis system response time (see paragraph (c) of this section). If the option described in paragraph (c)(2) of this section is used, a response time check is not required.
(i) Any location within the analysis system where a vacuum leak could affect the test results must be checked.
(ii) The maximum allowable leakage rate on the vacuum side is 0.5 percent of the in-use flow rate for the portion of the system being checked. the analyzer flows and bypass flows may be used to estimate the in-use flow rates.
(iii) The sample probe and the connection between the sample probe and valve V2 may be excluded from the leak check.
(ii) Option: If the flow rate for each flow meter is equal to or greater than the flow rate recorded in paragraph (c)(2)(i) of this section, then a pressure side leak check is not required.
(1) After any major change in the system, check the system response time by the following procedure:
(ii) Introduce an HC span gas into the sampling system at the sample probe or valve V2 at atmospheric pressure. Simultaneously, start the time measurement.
(iii) When the HC instrument response is 95 percent of the span gas concentration used, stop the time measurement.
(2) Option. If the following parameters are determined, the initial system response time may be generally applied to future checks:
(A) Determine by experimentation the minimum analyzer and bypass flow rates individually and in combination that will produce a response time as close as possible to 20.0 seconds per paragraph (c)(1) of this section.
(B) Record the highest minimum flow rate for each flow meter as determined in paragraph (c)(2)(i)(A) of this section.
(ii) Capillary flow analyzers. This procedure is applicable only to analyzers that have sample capillaries such as the HFID and CL analyzers. It is also assumed that the system has sample/span valves that perform the function of valves V9 and V13 in.
(B) Adjust the bypass flow rate to the flow rate recorded in paragraph (c)(2)(i)(B) of this section.
(C) Measure and record the response time from the sample/span valve(s) per paragraph (c)(1) of this section.
(D) The response time required by paragraph (c)(2)(ii)(C) of this section can be determined by switching from the “sample” position to the “span” position of the sample/span valve and observing the analyzer response on a chart recorder. Normally, the “sample” position would select a “room air” sample and the “span” position would select a span gas.
(F) Measure and record the response time from the sample/span valve(s) per paragraph (c)(1) of this section.
Title 40 published on 2014-07-01
no entries appear in the Federal Register after this date.