(a) A static-flow test will be performed on all open-circuit apparatus.
(b) The flow from the apparatus shall be greater than 200 liters per minute when the pressure in the facepiece of demand-apparatus is lowered by 51 mm. (2 inches) water-column height when full container pressure is applied.
(c) Where pressure demand apparatus are tested, the flow will be measured at zero gage pressure in the facepiece.
(d) Where apparatus with compressed-breathing-gas containers are tested, the flow test shall also be made with 3,450 kN/m.2 (500 p.s.i.g.) container pressure applied.
Title 42 published on 2013-10-01
no entries appear in the Federal Register after this date.
This is a list of United States Code sections, Statutes at Large, Public Laws, and Presidential Documents, which provide rulemaking authority for this CFR Part.