30 CFR 250.1203 - Gas measurement.
(a)To which meters do BSEE requirements for gas measurement apply? BSEE requirements for gas measurements apply to all OCS gas royalty and allocation meters.
(b)What are the requirements for measuring gas? You must:
(1) Submit a written application to, and obtain approval from, the Regional Supervisor before commencing gas production, or making any changes to the previously-approved measurement and/or allocation procedures. Your application (which may also include any relevant liquid hydrocarbon measurement and surface commingling requests) must be accompanied by payment of the service fee listed in § 250.125. The service fees are divided into two levels based on complexity, see table in § 250.1202(a)(1).
(2) Design, install, use, maintain, and test measurement equipment and procedures to ensure accurate and verifiable measurement. You must follow the recommendations in API MPMS or RP and AGA as incorporated by reference in 30 CFR 250.198, including the following additional editions:
(i) API RP 86 (incorporated by reference as specified in § 250.198);
(ii) AGA Report No. 7 (incorporated by reference as specified in § 250.198);
(iii) AGA Report No. 9 (incorporated by reference as specified in § 250.198);
(iv) AGA Report No. 10 (incorporated by reference as specified in § 250.198);
(3) Ensure that the measurement components demonstrate consistent levels of accuracy throughout the system.
(5) Take proportional-to-flow or spot samples upstream or downstream of the meter at least once every 6 months.
(6) When requested by the Regional Supervisor, provide available information on the gas quality.
(7) Ensure that standard conditions for reporting gross heating value (Btu) are at a base temperature of 60 °F and at a base pressure of 14.73 psia and reflect the same degree of water saturation as in the gas volume.
(8) When requested by the Regional Supervisor, submit copies of gas volume statements for each requested gas meter. Show whether gas volumes and gross Btu heating values are reported at saturated or unsaturated conditions; and
(c)What are the requirements for gas meter calibrations? You must:
(1) Verify/calibrate operating meters monthly, but do not exceed 42 days between verifications/calibrations. When a force majeure event precludes the required monthly meter verification/calibration, meters must be verified/calibrated within 15 days after being returned to service. The meters must be verified/calibrated monthly thereafter, but do not exceed 42 days between meter verifications/calibrations;
(2) Calibrate each meter by using the manufacturer's specifications;
(3) Conduct calibrations as close as possible to the average hourly rate of flow since the last calibration;
(4) Retain calibration reports at the field location for 2 years, and send the reports to the Regional Supervisor upon request; and
(5) Permit BSEE representatives to witness calibrations.
(d)What must I do if a gas meter is out of calibration or malfunctioning? If a gas meter is out of calibration or malfunctioning, you must:
(1) If the readings are greater than the contractual tolerances, adjust the meter to function properly or remove it from service and replace it.
(2) Correct the volumes to the last acceptable calibration as follows:
(i) If the duration of the error can be determined, calculate the volume adjustment for that period.
(ii) If the duration of the error cannot be determined, apply the volume adjustment to one-half of the time elapsed since the last calibration or 21 days, whichever is less.
(e)What are the requirements when natural gas from a Federal lease on the OCS is transferred to a gas plant before royalty determination? If natural gas from a Federal lease on the OCS is transferred to a gas plant before royalty determination:
(i) A copy of the monthly gas processing plant allocation statement; and
(ii) Gross heating values of the inlet and residue streams when not reported on the gas plant statement.
(2) If you measure the volume, document the measurement equipment used and include the volume measured.
(4) You must keep the documentation, including the volume data, easily obtainable for inspection at the field location for at least 2 years, and must retain the documentation at a location of your choosing for at least 7 years after the documentation is generated, subject to all other document retention and production requirements in 30 U.S.C. 1713 and 30 30 CFR part 1212.
Title 30 published on 06-Mar-2018 03:27
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 30 CFR Part 250 after this date.