(a) Gases for the CO and CO2 analyzers shall be single blends of CO and CO2. respectively, using nitrogen as the diluent.
(b) Gases for the hydrocarbon analyzer shall be:
(1) Single blends of propane using air as the diluent; and
(2) Optionally, for response factor determination, single blends of methanol using air as the diluent.
(c) Gases for the methane analyzer shall be single blends of methane using air as the diluent.
(d) Gases for the NOX analyzer shall be single blends of NO named as NOX with a maximum NO2 concentration of five percent of the nominal value using nitrogen as the diluent.
(e) Fuel for FIDs and HFIDs and methane analyzers shall be a blend of 40 ±2 percent hydrogen with the balance being helium. The mixture shall contain less than 1 ppm equivalent carbon response. 98 to 100 percent hydrogen fuel may be used with advance approval by the Administrator.
(f) The allowable zero gas (air or nitrogen) impurity concentrations shall not exceed 1 ppm equivalent carbon response, 1 ppm carbon monoxide, 0.04 percent (400 ppm) carbon dioxide and 0.1 ppm nitric oxide.
(1) “Zero-grade air” includes artificial “air” consisting of a blend of nitrogen and oxygen with oxygen concentrations between 18 and 21 mole percent.
(2) Calibration gases (not including methanol) shall be traceable to within one percent of NIST (formerly NBS) gas standards, or other gas standards which have been approved by the Administrator.
(3) Span gases (not including methanol) shall be accurate to within two percent of true concentration, where true concentration refers to NIST (formerly NBS) gas standards, or other gas standards which have been approved by the Administrator.
(4) Methanol in air gases used for response factor determination shall:
(i) Be traceable to within ±2 percent of NIST (formerly NBS) gas standards, or other standards which have been approved by the Administrator; and
(ii) Remain within ±2 percent of the labeled concentration. Demonstration of stability shall be based on a quarterly measurement procedure with a precision of ±2 percent (two standard deviations), or other method approved by the Administrator. The measurement procedure may incorporate multiple measurements. If the true concentration of the gas changes by more than two percent, but less than ten percent, the gas may be relabeled with the new concentration.
(h) The use of precision blending devices (gas dividers) to obtain the required calibration gas concentrations is acceptable, provided that the blended gases are accurate to within ±1.5 percent of NBS gas standards, or other gas standards which have been approved by the Administrator. This accuracy implies that primary gases used for blending must be “named” to an accuracy of at least ±1 percent, traceable to NBS or other approved gas standards.
[59 FR 48530, Sept. 21, 1994, as amended at 60 FR 34371, June 30, 1995]
Beta! The text on the eCFR tab represents the unofficial eCFR text at ecfr.gov.
This section’s status may have changed. It may have been renumbered, reserved, or removed.
This is a list of United States Code sections, Statutes at Large, Public Laws, and Presidential Documents, which provide rulemaking authority for this CFR Part.