40 CFR Part 53, Subpart C, Table C-3 to Subpart C of Part 53 - Test Specifications for Pb in TSP and Pb in PM 10 Methods
Table C-3 to Subpart C of Part 53—Test Specifications for Pb in TSP and Pb in PM 10 Methods
|Concentration range equivalent to percentage of NAAQS in µg/m3||30% to 250%|
|Minimum number of 24-hr measurements||5|
|Maximum reference method analytical bias, Dq||±5%|
|Maximum precision, PR or PC||≤15%|
|Maximum difference (D)||±20%|
|Estimated Method Detection Limit (MDL), µg/m3||5% of NAAQS level.|
[73 FR 67059, Nov. 12, 2008]
Title 40 published on 2015-07-01.
No entries appear in the Federal Register after this date, for 40 CFR Part 53.