46 CFR 162.050-19 - Oil content meter and bilge alarm test rig.
(a) This section contains requirements for test rigs used in approval testing of oil content meters and meter. A typical test rig is described in Figure 162.050-19. The mixture pipe shown in Figure 162.050-19 is the portion of test rig piping between the oil injection point and the meter or bilge alarm piping.
(b) Each sample point on a test rig must be of the type described in Figure 162.050-17(e) and must be in a vertical portion of the test rig piping.
(c) Each test rig must have a centrifugal pump that is designed to operate at 1,000 revolutions per minute or more.
(d) The mixture pipe on a test rig must have a uniform inside diameter.
Title 46 published on 09-May-2017 04:27
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 46 CFR Part 162 after this date.