Or. Admin. R. 333-106-0320 - Radiation from Capacitor Energy Storage Equipment in Standby Status
Current through Register Vol. 61, No. 4, April 1, 2022
Radiation emitted from the X-ray tube when the exposure switch or timer is not activated shall not exceed a rate of two milliroentgens (0.516 uC/kg) per hour at five centimeters from any accessible surface of the diagnostic source assembly, with the beam-limiting device fully open.
Stat. Auth.: ORS 453.605 - 453.755
Stats. Implemented: ORS 453.605 - 453.755
The following state regulations pages link to this page.
State regulations are updated quarterly; we currently have two versions available. Below is a comparison between our most recent version and the prior quarterly release. More comparison features will be added as we have more versions to compare.