Tenn. Comp. R. & Regs. 0400-20-07-.78 - THERAPY-RELATED COMPUTER SYSTEMS
(1) A licensee shall perform or shall verify and maintain documentation of acceptance testing on the treatment planning system in accordance with published protocols accepted by nationally recognized bodies. At a minimum, the acceptance testing must include, as applicable, verification of:
(a) The source-specific input parameters required by the dose calculation algorithm;
(b) The accuracy of dose, dwell time, and treatment time calculations at representative points;
(c) The accuracy of isodose plots and graphic displays;
(d) The accuracy of the software used to determine sealed source positions from radiographic images; and
(e) The accuracy of electronic transfer of the treatment delivery parameters to the treatment delivery unit from the treatment planning system.
Authority: T.C.A. §§ 68-202-101 et seq., 68-202-201 et seq. and 4-5-201 et seq.
State regulations are updated quarterly; we currently have two versions available. Below is a comparison between our most recent version and the prior quarterly release. More comparison features will be added as we have more versions to compare.
No prior version found.