21 CFR 820.30 - Design controls.
(1) Each manufacturer of any class III or class II device, and the class I devices listed in paragraph (a)(2) of this section, shallestablish and maintain procedures to control the design of the device in order to ensure that specified design requirements are met.
(2) The following class I devices are subject to design controls:
(i) Devices automated with computer software; and
(ii) The devices listed in the following chart.
|868.6810||Catheter, Tracheobronchial Suction.|
|892.5650||System, Applicator, Radionuclide, Manual.|
|892.5740||Source, Radionuclide Teletherapy.|
(b)Design and development planning. Each manufacturershallestablish and maintain plans that describe or reference the design and development activities and define responsibility for implementation. The plans shall identify and describe the interfaces with different groups or activities that provide, or result in, input to the design and development process. The plans shall be reviewed, updated, and approved as design and development evolves.
(c)Design input. Each manufacturershallestablish and maintain procedures to ensure that the design requirements relating to a device are appropriate and address the intended use of the device, including the needs of the user and patient. The procedures shall include a mechanism for addressing incomplete, ambiguous, or conflicting requirements. The design input requirements shall be documented and shall be reviewed and approved by a designated individual(s). The approval, including the date and signature of the individual(s) approving the requirements, shall be documented.
(d)Design output. Each manufacturershallestablish and maintain procedures for defining and documenting design output in terms that allow an adequate evaluation of conformance to design input requirements. Design output procedures shall contain or make reference to acceptance criteria and shall ensure that those design outputs that are essential for the proper functioning of the device are identified. Design outputshall be documented, reviewed, and approved before release. The approval, including the date and signature of the individual(s) approving the output, shall be documented.
(e)Design review. Each manufacturershallestablish and maintain procedures to ensure that formal documented reviews of the design results are planned and conducted at appropriate stages of the device's design development. The procedures shall ensure that participants at each design review include representatives of all functions concerned with the design stage being reviewed and an individual(s) who does not have direct responsibility for the design stage being reviewed, as well as any specialists needed. The results of a design review, including identification of the design, the date, and the individual(s) performing the review, shall be documented in the design history file (the DHF).
(f)Design verification. Each manufacturershallestablish and maintain procedures for verifying the device design. Design verification shall confirm that the design output meets the design input requirements. The results of the design verification, including identification of the design, method(s), the date, and the individual(s) performing the verification, shall be documented in the DHF.
(g)Design validation. Each manufacturershallestablish and maintain procedures for validating the device design. Design validationshall be performed under defined operating conditions on initial production units, lots, or batches, or their equivalents. Design validationshall ensure that devices conform to defined user needs and intended uses and shall include testing of production units under actual or simulated use conditions. Design validationshall include software validation and risk analysis, where appropriate. The results of the design validation, including identification of the design, method(s), the date, and the individual(s) performing the validation, shall be documented in the DHF.
(i)Design changes. Each manufacturershallestablish and maintain procedures for the identification, documentation, validation or where appropriate verification, review, and approval of design changes before their implementation.
(j)Design history file. Each manufacturershallestablish and maintain a DHF for each type of device. The DHF shall contain or reference the records necessary to demonstrate that the design was developed in accordance with the approved design plan and the requirements of this part.
Title 21 published on 2015-12-03
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 21 CFR Part 820 after this date.
- 21 CFR 4.4 — How Can I Comply With These Current Good Manufacturing Practice Requirements for a Co-Packaged or Single-Entity Combination Product?
- 21 CFR 812.35 — Supplemental Applications.
- 21 CFR 801.45 — Devices That Must Be Directly Marked With a Unique Device Identifier.
- 21 CFR 812.1 — Scope.
- 21 CFR 820.1 — Scope.
Title 21 published on 2015-12-03.
No entries appear in the Federal Register after this date, for 21 CFR Part 820.