42 CFR 494.60 - Condition: Physical environment.
The dialysis facility must be designed, constructed, equipped, and maintained to provide dialysis patients, staff, and the public a safe, functional, and comfortable treatment environment.
(a)Standard: Building. The building in which dialysis services are furnished must be constructed and maintained to ensure the safety of the patients, the staff, and the public.
(b)Standard: Equipment maintenance. The dialysis facility must implement and maintain a program to ensure that all equipment (including emergency equipment, dialysis machines and equipment, and the water treatment system) are maintained and operated in accordance with the manufacturer's recommendations.
(c)Standard: Patient care environment.
(2) The dialysis facility must:
(i) Maintain a comfortable temperature within the facility; and
(ii) Make reasonable accommodations for the patients who are not comfortable at this temperature.
(d)Standard: Fire safety.
(1) Except as provided in paragraph (e)(2) of this section, by February 9, 2009, dialysis facilities that are located adjacent to high hazardous occupancies or do not provide one or more exits to the outside at grade level from the patient treatment area level, must comply with applicable provisions of the 2000 edition of the Life Safety Code of the National Fire Protection Association (which is incorporated by reference at § 403.744(a)(1)(i) of this chapter).
(2) Notwithstanding paragraph (e)(1) of this section, dialysis facilities participating in Medicare as of October 14, 2008 that require sprinkler systems are those housed in multi-story buildings construction Types II(000), III(200), or V(000), as defined in the 2000 edition of the Life Safety Code of the National Fire Protection Association (which is incorporated by reference at § 403.744(a)(1)(i) of this chapter), section 220.127.116.11, which were constructed after January 1, 2008, and those housed in high rise buildings over 75 feet in height, which were constructed after January 1, 2008.
(3) If CMS finds that a fire and safety code imposed by the facility's State law adequately protects a dialysis facility's patients, CMS may allow the State survey agency to apply the State's fire and safety code instead of the Life Safety Code.
(4) After consideration of State survey agency recommendations, CMS may waive, for individual dialysis facilities and for appropriate periods, specific provisions of the Life Safety Code, if the following requirements are met:
(i) The waiver would not adversely affect the health and safety of the dialysis facility's patients; and
(ii) Rigid application of specific provisions of the Life Safety Code would result in an unreasonable hardship for the dialysis facility.
Title 42 published on 19-Apr-2017 03:51
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 42 CFR Part 494 after this date.