40 CFR 60.485 - Test methods and procedures.
(a) In conducting the performance tests required in § 60.8, the owner or operator shall use as reference methods and procedures the test methods in appendix A of this part or other methods and procedures as specified in this section, except as provided in § 60.8(b).
(1) Method 21 shall be used to determine the presence of leaking sources. The instrument shall be calibrated before use each day of its use by the procedures specified in Method 21. The following calibration gases shall be used:
(i) Zero air (less than 10 ppm of hydrocarbon in air); and
(ii) A mixture of methane or n-hexane and air at a concentration of about, but less than, 10,000 ppm methane or n-hexane.
(1) The requirements of paragraph (b) shall apply.
(2) Method 21 shall be used to determine the background level. All potential leak interfaces shall be traversed as close to the interface as possible. The arithmetic difference between the maximum concentration indicated by the instrument and the background level is compared with 500 ppm for determining compliance.
(d) The owner or operator shall test each piece of equipment unless he demonstrates that a process unit is not in VOC service, i.e., that the VOC content would never be reasonably expected to exceed 10 percent by weight. For purposes of this demonstration, the following methods and procedures shall be used:
(1) Procedures that conform to the general methods in ASTM E 260-73, 91, or 96, E 168-67, 77, or 92, E 169-63, 77, or 93 (incorporated by reference - see § 60.17) shall be used to determine the percent VOC content in the process fluid that is contained in or contacts a piece of equipment.
(2) Organic compounds that are considered by the Administrator to have negligible photochemical reactivity may be excluded from the total quantity of organic compounds in determining the VOC content of the process fluid.
(3) Engineering judgment may be used to estimate the VOC content, if a piece of equipment had not been shown previously to be in service. If the Administrator disagrees with the judgment, paragraphs (d) (1) and (2) of this section shall be used to resolve the disagreement.
(1) The vapor pressure of one or more of the organic components is greater than 0.3 kPa at 20 °C (1.2 in. H2O at 68 °F). Standard reference texts or ASTM D 2879-83, 96, or 97 (incorporated by reference - see § 60.17) shall be used to determine the vapor pressures.
(2) The total concentration of the pure organic components having a vapor pressure greater than 0.3 kPa at 20 °C (1.2 in. H2O at 68 °F) is equal to or greater than 20 percent by weight.
(3) The fluid is a liquid at operating conditions.
(f) Samples used in conjunction with paragraphs (d), (e), and (g) of this section shall be representative of the process fluid that is contained in or contacts the equipment or the gas being combusted in the flare.
(1) Method 22 shall be used to determine visible emissions.
(2) A thermocouple or any other equivalent device shall be used to monitor the presence of a pilot flame in the flare.
(3) The maximum permitted velocity for air assisted flares shall be computed using the following equation:
(4) The net heating value (HT) of the gas being combusted in a flare shall be computed using the following equation:
(5) Method 18 or ASTM D 6420-99 (2004) (where the target compound(s) are those listed in Section 1.1 of ASTM D 6420-99, and the target concentration is between 150 parts per billion by volume and 100 parts per million by volume) and ASTM D 2504-67, 77 or 88 (Reapproved 1993) (incorporated by reference - see § 60.17) shall be used to determine the concentration of sample component “i.”
(6) ASTM D 2382-76 or 88 or D 4809-95 (incorporated by reference - see § 60.17) shall be used to determine the net heat of combustion of component “i” if published values are not available or cannot be calculated.
(7) Method 2, 2A, 2C, or 2D, as appropriate, shall be used to determine the actual exit velocity of a flare. If needed, the unobstructed (free) cross-sectional area of the flare tip shall be used.
(1) The percent of valves leaking shall be determined using the following equation:
(2) The total number of valves monitored shall include difficult-to-monitor and unsafe-to-monitor valves only during the monitoring period in which those valves are monitored.
(3) The number of valves leaking shall include valves for which repair has been delayed.
(4) Any new valve that is not monitored within 30 days of being placed in service shall be included in the number of valves leaking and the total number of valves monitored for the monitoring period in which the valve is placed in service.
(6) The total number of valves monitored does not include a valve monitored to verify repair.
Title 40 published on 2015-08-22
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 40 CFR Part 60 after this date.
- 40 CFR 60.633 — Exceptions.
- 40 CFR 60.483-2 — Alternative Standards for Valves - Skip Period Leak Detection and Repair.
- 40 CFR 60.481 — Definitions.
- 40 CFR 60.5400 — What Equipment Leak Standards Apply to Affected Facilities at an Onshore Natural Gas Processing Plant?
- 40 CFR 60.562-2 — Standards: Equipment Leaks of VOC.
- 40 CFR 60.482-7 — Standards: Valves in Gas/Vapor Service and in Light Liquid Service.
- 40 CFR 60.483-1 — Alternative Standards for Valves - Allowable Percentage of Valves Leaking.
- 40 CFR 60.593 — Exceptions.
- 40 CFR 60.486 — Recordkeeping Requirements.
- 40 CFR 63.161 — Definitions.
- 40 CFR 60.482-4 — Standards: Pressure Relief Devices in Gas/Vapor Service.
- 40 CFR 60.592 — Standards.
- 40 CFR 63.648 — Equipment Leak Standards.
- 40 CFR 60.482-8 — Standards: Pumps and Valves in Heavy Liquid Service, Pressure Relief Devices in Light Liquid or Heavy Liquid Service, and Connectors.
- 40 CFR 60.482-10 — Standards: Closed Vent Systems and Control Devices.
- 40 CFR 60.632 — Standards.
- 40 CFR 60.482-3 — Standards: Compressors.
- 40 CFR 63.1361 — Definitions.
- 40 CFR 60.631 — Definitions.
- 40 CFR 60.112b — Standard for Volatile Organic Compounds (VOC).
- 40 CFR 60.482-2 — Standards: Pumps in Light Liquid Service.