40 CFR 60.485a - Test methods and procedures.
(a) In conducting the performance tests required in § 60.8, the owner or operator shall use as reference methods and procedures the test methods in appendix A of this part or other methods and procedures as specified in this section, except as provided in § 60.8(b).
(1) Method 21 shall be used to determine the presence of leaking sources. The instrument shall be calibrated before use each day of its use by the procedures specified in Method 21 of appendix A-7 of this part. The following calibration gases shall be used:
(i) Zero air (less than 10 ppm of hydrocarbon in air); and
(ii) A mixture of methane or n-hexane and air at a concentration no more than 2,000 ppm greater than the leak definition concentration of the equipment monitored. If the monitoring instrument's design allows for multiple calibration scales, then the lower scale shall be calibrated with a calibration gas that is no higher than 2,000 ppm above the concentration specified as a leak, and the highest scale shall be calibrated with a calibration gas that is approximately equal to 10,000 ppm. If only one scale on an instrument will be used during monitoring, the owner or operator need not calibrate the scales that will not be used during that day's monitoring.
(2) A calibration drift assessment shall be performed, at a minimum, at the end of each monitoring day. Check the instrument using the same calibration gas(es) that were used to calibrate the instrument before use. Follow the procedures specified in Method 21 of appendix A-7 of this part, Section 10.1, except do not adjust the meter readout to correspond to the calibration gas value. Record the instrument reading for each scale used as specified in § 60.486a(e)(7). Calculate the average algebraic difference between the three meter readings and the most recent calibration value. Divide this algebraic difference by the initial calibration value and multiply by 100 to express the calibration drift as a percentage. If any calibration drift assessment shows a negative drift of more than 10 percent from the initial calibration value, then all equipment monitored since the last calibration with instrument readings below the appropriate leak definition and above the leak definition multiplied by (100 minus the percent of negative drift/divided by 100) must be re-monitored. If any calibration drift assessment shows a positive drift of more than 10 percent from the initial calibration value, then, at the owner/operator's discretion, all equipment since the last calibration with instrument readings above the appropriate leak definition and below the leak definition multiplied by (100 plus the percent of positive drift/divided by 100) may be re-monitored.
(1) The requirements of paragraph (b) shall apply.
(2) Method 21 of appendix A-7 of this part shall be used to determine the background level. All potential leak interfaces shall be traversed as close to the interface as possible. The arithmetic difference between the maximum concentration indicated by the instrument and the background level is compared with 500 ppm for determining compliance.
(d) The owner or operator shall test each piece of equipment unless he demonstrates that a process unit is not in VOC service, i.e., that the VOC content would never be reasonably expected to exceed 10 percent by weight. For purposes of this demonstration, the following methods and procedures shall be used:
(1) Procedures that conform to the general methods in ASTM E 260-73, 91, or 96, E 168-67, 77, or 92, E 169-63, 77, or 93 (incorporated by reference - see § 60.17) shall be used to determine the percent VOC content in the process fluid that is contained in or contacts a piece of equipment.
(2) Organic compounds that are considered by the Administrator to have negligible photochemical reactivity may be excluded from the total quantity of organic compounds in determining the VOC content of the process fluid.
(3) Engineering judgment may be used to estimate the VOC content, if a piece of equipment had not been shown previously to be in service. If the Administrator disagrees with the judgment, paragraphs (d)(1) and (2) of this section shall be used to resolve the disagreement.
(1) The vapor pressure of one or more of the organic components is greater than 0.3 kPa at 20 °C (1.2 in. H2O at 68 °F). Standard reference texts or ASTM D 2879-83, 96, or 97 (incorporated by reference - see § 60.17) shall be used to determine the vapor pressures.
(2) The total concentration of the pure organic components having a vapor pressure greater than 0.3 kPa at 20 °C (1.2 in. H2O at 68 °F) is equal to or greater than 20 percent by weight.
(3) The fluid is a liquid at operating conditions.
(f) Samples used in conjunction with paragraphs (d), (e), and (g) of this section shall be representative of the process fluid that is contained in or contacts the equipment or the gas being combusted in the flare.
(1) Method 22 of appendix A-7 of this part shall be used to determine visible emissions.
(2) A thermocouple or any other equivalent device shall be used to monitor the presence of a pilot flame in the flare.
(3) The maximum permitted velocity for air assisted flares shall be computed using the following equation:
(4) The net heating value (HT) of the gas being combusted in a flare shall be computed using the following equation:
(5) Method 18 of appendix A-6 of this part or ASTM D 6420-99 (2004) (where the target compound(s) are those listed in Section 1.1 of ASTM D 6420-99, and the target concentration is between 150 parts per billion by volume and 100 ppmv) and ASTM D 2504-67, 77, or 88 (Reapproved 1993) (incorporated by reference-see § 60.17) shall be used to determine the concentration of sample component “i.”
(6) ASTM D 2382-76 or 88 or D 4809-95 (incorporated by reference-see § 60.17) shall be used to determine the net heat of combustion of component “i” if published values are not available or cannot be calculated.
(7) Method 2, 2A, 2C, or 2D of appendix A-7 of this part, as appropriate, shall be used to determine the actual exit velocity of a flare. If needed, the unobstructed (free) cross-sectional area of the flare tip shall be used.
(1) The percent of valves leaking shall be determined using the following equation:
(2) The total number of valves monitored shall include difficult-to-monitor and unsafe-to-monitor valves only during the monitoring period in which those valves are monitored.
(3) The number of valves leaking shall include valves for which repair has been delayed.
(4) Any new valve that is not monitored within 30 days of being placed in service shall be included in the number of valves leaking and the total number of valves monitored for the monitoring period in which the valve is placed in service.
(6) The total number of valves monitored does not include a valve monitored to verify repair.
Title 40 published on 2015-08-22
The following are ALL rules, proposed rules, and notices (chronologically) published in the Federal Register relating to 40 CFR Part 60 after this date.
- 40 CFR 60.481a — Definitions.
- 40 CFR 60.5401 — What Are the Exceptions to the Equipment Leak Standards for Affected Facilities at Onshore Natural Gas Processing Plants?
- 40 CFR 60.5400 — What Equipment Leak Standards Apply to Affected Facilities at an Onshore Natural Gas Processing Plant?
- 40 CFR 60.592a — Standards.
- 40 CFR 60.5430 — What Definitions Apply to This Subpart?
- 40 CFR 60.482-7a — Standards: Valves in Gas/Vapor Service and in Light Liquid Service.
- 40 CFR 60.593a — Exceptions.
- 40 CFR 60.486a — Recordkeeping Requirements.
- 40 CFR 60.482-4a — Standards: Pressure Relief Devices in Gas/Vapor Service.
- 40 CFR 60.483-2a — Alternative Standards for Valves - Skip Period Leak Detection and Repair.
- 40 CFR 60.482-2a — Standards: Pumps in Light Liquid Service.
- 40 CFR 60.482-8a — Standards: Pumps, Valves, and Connectors in Heavy Liquid Service and Pressure Relief Devices in Light Liquid or Heavy Liquid Service.
- 40 CFR 60.482-3a — Standards: Compressors.
- 40 CFR 60.482-10a — Standards: Closed Vent Systems and Control Devices.
- 40 CFR 60.483-1a — Alternative Standards for Valves - Allowable Percentage of Valves Leaking.
- 40 CFR 60.482-11a — Standards: Connectors in Gas/Vapor Service and in Light Liquid Service.
- 40 CFR 60.5401a — What Are the Exceptions to the Equipment Leak GHG and VOC Standards for Affected Facilities at Onshore Natural Gas Processing Plants?
- 40 CFR 60.5430a — What Definitions Apply to This Subpart?
- 40 CFR 60.5400a — What Equipment Leak GHG and VOC Standards Apply to Affected Facilities at an Onshore Natural Gas Processing Plant?